The HP B is a single channel W DC programmable power supply with GPIB and RS connectivity. It is part of the HP x series. Agilent B/B/B/B System DC Power. Supplies All references to HP-IB have been changed to GPIB. A note about. I just picked up two HP B power supplies as I read somewhere quickly that it not only can be programmed via GPIB to set voltage/current.
|Published (Last):||7 June 2009|
|PDF File Size:||4.63 Mb|
|ePub File Size:||8.15 Mb|
|Price:||Free* [*Free Regsitration Required]|
Enter the new password from the keypad.
After some head scratching and a LANG change this was resolved. Page 50 A – SpecificationsSupplemental CharacteristicsTable A-2 lists the supplemental characteristics, which are not warranted 6632h are descriptions oftypical performance determined either by design or type testing.
ComiXology Thousands of Digital Comics. In addition to letting me get at the DIP switches, this also gave me a spiritual sense of wellness at having un-funked a nice piece of gear that somebody had really let go. After much swearing, compressed air, and windex, the inside of the chassis looked like this: The Shift annunciator is lit,indicating that the shifted keys are active.
Showing Results for Keysight Technologies (Agilent HP) 6632B
ProtIndicates that the output has been disabled by one of theprotection features. This chapter provides a preliminary introduction to the dc source front panel. CopyleftAll Wrongs Reserved.
Page 22 3 – InstallationLocationThe outline diagram in figure gives the dimensions of your dc source. The dc source must beinstalled in a location that allows sufficient space at the sides and back of the unit for adequate aircirculation see Bench Operation.
V or A MAX One of the cool things about these supplies is that they talk RS in many languages. Although my original post about this is pretty old I still use this same unit as my main bench supply. Page 56 B – Verification and Calibration5. Page 75 Manual UpdatesThe following updates have been made to this manual since the November printingindicated on the Printing History page.
Save this state to location 1. The dc source must pass turn-on selftest before you can proceed with theverification tests.
Agilent / HP B Watt DC System Power Supply 20v 5a | eBay
Selftest ErrorsPressing the Shift, Error keys will show the error number. Enter the existing password from Entry keypad and press Enter3. OCP The overcurrent protection state is enabled. Front Panel Operation – 5Entry KeysRefer to the examples jp in this chapter for more details on the use of these keys.
Unscrew the line fuse cap from the rear panel and verify that the fuse rating matches what isspecified on the FUSES label on the rear panel. UnrThe output is unregulated. 6632b Choose a language for shopping. Verification and Calibration – BTable B- 3.
After much swearing, compressed air, and windex, the inside of the chassis looked like this:. The next problem was the fact that the front panel meter was a little off from what I was measuring at the output. The calibration procedure can only be performed using the SCPI languagecommands. Amazon Second Chance Pass it on, trade it in, give it a second life.
If looking through the HP Hewlett-Packard B user manual directly on this website is not convenient for you, there are two possible solutions:.
When the unit is operating in frontpanel meter mode, all front panel measurements are calculated from np total of readings takenat a Instructions are given for performing the procedures eitherfrom the front panel or from a controller over the HP-IB. Yesterday Narrat1ve got a new sort of bench power supply in the form of an eBayed HP b!
Page 18 2 — General InformationSafety ConsiderationsThis dc source is a Safety Class 1 instrument, which means it has a protective earth terminal. Still, if your device is significantly out of cal, you can get a lot closer. This thing is heavy!! Parameter lists are circular; you can return to the startingposition by continuously pressing either key.
Enter the calibration password from Entry keypad and press Enter. Seechapter 5 for more details.
Page 31 4Turn-On CheckoutIntroductionSuccessful tests in this chapter provide a high degree of confidence that the dc source is operatingproperly. It also made me hl worried that there were secret vermin in my apartment that were taking refuge in my bench. Anyhoo, the unit showed up all kinds of Dirt McGirt: