Keyboard Layout v Fluke B/1 96B-C/B-C Service Manual List of Figures For the mA. current D pin 3 is connected to pin 1, and pin 13 is. Manual de Servicio Fluke b – Ebook download as PDF File .pdf), Text File . txt) or trigger level mV) or (R the switches in D are all open. F HP Deskjet F HP Deskjet D HP Deskjet D HP Deskjet D . would work while others would not, and maybe it’s some strange fluke.
|Published (Last):||22 December 2012|
|PDF File Size:||9.24 Mb|
|ePub File Size:||2.8 Mb|
|Price:||Free* [*Free Regsitration Required]|
You will hear a very weak beep now. If one or more currents OK then check D and control signals. Connect a dc ampere fluek between the banana jack inputs. There’s a problem loading this menu right now. Save the Calibration Data and Exit the calibration mode, section 5. Replay modes Step by Step, Replay as Animation 2. When you are finished, set the A to 0 zero Volt, and to Standby.
Trigger Path See Figure Periodically dynamic calibration will cause bursts to be shown if no key is pressed. Correct versions added Remote control: Install the keypad support plate item 6. The display will show WarmingUp CL This indicates that the communication between the Terminal program and the test tool is accomplished.
Voltage measurements The input voltage is attenuated by a factor 4, 40, or Trace disturbance with VPS probe shorted Table The D-ASIC acquires the digital equivalents of the samples to process them and show them on the display as traces and readings.
Install the battery pack, and re-attach the cable. Display Control The displayed screen consists of: MS near to the LCD and keypad foil connectors.
Remove the keypad support plate item 5. Due to calibrator noise, occasionally OL overload can be shown. Keep the lens free of dust and grease.
HP C9352AE (22) Multi Colour Cartridge for Inkjet Printer
Press to start the zero calibration 5. Loosen 2 Input Cover Screws Now, barely 3 weeks later, they are both basically completely out and it is time for more refills.
S-ASIC signal section block diagram One-Way List Drawing Tool Press a key, and check ROW Connect the test tool to the calibrator as shown in Figure IDLE invalidand the softkey menu. Because of the highly integrated design of the test tool, it is not always necessary to check all features separately. If correct, the mask software runs, but the buzzer circuit does not function. Remove the display assembly item 6. Calibrate the display contrast see section 5.
Full text of “FLUKE B, B, C, B,C Service”
Connections between V4 1 02 and N Note 2 If you press NOthe test tool returns to the calibration mode. Error span fixed Volume: Taking it into use As the module is not repairable, no detailed description and diagrams are provided. If wrong, this is possibly caused by V, R, short to ground, loose pins of N, or N defective. Starting Fault Finding Proceed as follows to test the Input A and B automatic scope ac Voltage measurement accuracy: Set the A to operate OPR. With this technology, printing is smoother and more consistent.
Remove all test leads from the test tool inputs. PCA revision number sticker Axdia reserves the right to modify product offerings or specifications without advance notice. Calibrate the display contrast see section 5.
Configuration Requirements 3 3 Software Installation Note 2 If you press ftllfcl NO ,the test tool returns to the calibration mode.
Floating Voltage from any terminal to ground. The more impressive part of my experience is this seller’s dedication to customer service. The output voltage of the backlight voltage rluke possibly can cause damage to the Main PCA when no load is connected for more than some minutes. Turn on 1-Click ordering for this browser. Now press Sfe’S Continue to start the Observe the 5 A output voltage and check to flukr if it is within the range shown under the appropriate column.