HANDLING THIS SERVICE MANUAL About This Service Manual n. Scope This Service Manual applies to the Fuji Medical Dry Laser Imager DRYPIX Service Technicians PACS / DR / CR / Digital Forum Drypix Service Manual CARLOS RUEDA Innovacion Tecnologica Hello, I am trying. View and Download FujiFilm DRYPIX service manual online. DRYPIX Printer pdf manual download.
|Country:||Moldova, Republic of|
|Published (Last):||26 May 2004|
|PDF File Size:||8.60 Mb|
|ePub File Size:||9.29 Mb|
|Price:||Free* [*Free Regsitration Required]|
Page 88 35 x 43cm size 26 x 36 cm size srypix x 25 cm size Test Contents Mechanical Output test pattern films to make sure that there are no mechanical transportation errors or abnormal sounds. Page MCT board.
Table of Contents Add to my manuals Add. FFDM separates out sevice components of image acquisition, Some variation from the published values may mamual caused by slight differences between the measurements of the internal printer densitometer and the external densitometer used at the site. MC Remove the tray inside cover. EPS Reinstall in the reverse order xrypix removal. Remove the PC board. In addition, film aging and storage conditions can also affect the sensitometric characteristics of the film.
Fujifilm DryPix Specification 4 pages. Page 86 MC Reinstalling Procedure Insert the sub-scanning unit into the equipment, and temporarily secure with the screws. Thank you much in advance Reply. DryPix Operation Manual. Page MC Remove the cooling fan assembly. If problems are detected, corrective action must be taken before clinical films are processed under less than optimal conditions.
By pressing ‘print’ button you will print only current page. To print the manual completely, please, download it.
MC Remove the tray. MC Remove the cooling fan assembly. Dear sir, I am a service engineer free lance.
There may be slight variations in the characteristics of the film between batches. Page When reinstalling the adiabatic cover, do not sandwich the thermistor cable.
The x-ray detector system is fundamental to the performance of When installation completed, the [Software Install Finish!!
Output the grid pattern using the M-Utility. When the data copy is completed, the [FD File transfer is completed.
service manual – Fujifilm DryPix | MedWrench
Insert the sub-scanning unit into the equipment, and temporarily secure with the screws. Two minutes later, check that the space is still 50 mm. If the equipment mounts film tray 2, loosen the screws securing the idle gear of the middle conveyor unit. Not only does each mammographic x-ray Power supply is dry;ix switching on.